CEA 10 error message: Overflow of measured value transmit buffer
Solution A description of this resolution:
- Check configuration and correctly adjust filtering
Log on via the menu item ¡¯Unit information¡¯ to the corresponding CEA 10 and select ¡®Read Confi. Data¡¯. The settings and channel assignments must be compared with those of the editor program. There may only be as many channel numbers present as the number of channels assigned in the CEA 10.
The following values should be set for each channel:
Non-linear filter NIFIL
Linear range LB
Integration time T1